Seta Multiflash Cleveland Flash Point Module – 34300-2
The automatic Cleveland module conforms precisely to national and international Cleveland open cup flash point test methods. It comprises a sample cup, temperature probe, flash/fire detector, and a DIPS pod containing the sweeping mechanism with co-axial gas/electric ignitor.
The Cleveland module must be used with the Multiflash Automatic Flash Point – Universal Base Unit (34000-0). The base unit recognizes that the Cleveland module is connected and instantaneously sets up standard test parameters and calibration data.
Methods: ASTM D92, IP 36, BS DIN EN ISO 2592
- Fully self-contained flash/fire point test module
- Ambient to 400 °C
- Gas or electric ignition
- Automatic heating control, flash and fire detection
- Automatic test results & end of test audible warning
- Automatic snuffer
- Automatic barometric correction
The flashpoint test area consists of a removable sample cup in a thermal insulation ring, and a probe block for mounting the flash/fire detector and sample temperature probe. The sample cup is heated by the hotplate within the Multiflash Automatic Flash Point - Universal Base Unit (34000-0).
The DIPS pod contains the ignitor sweeping mechanism with a coaxial gas jet/electric hot wire ignitor and a snuffer that extinguishes any flames in the sample cup at the end of the test.
The sample temperature probe is a class A PRT, and flash/fire detection is by an ionization ring.
Safety features include an initial sweep to check that the sample is below its flashpoint at the start of the test and a PRT check to ensure that the correct probe is connected and is immersed in the sample. When in search mode, the software limits the maximum temperature to 50 °C above "expected flash". The snuffer automatically deploys at the end of the test or if the independent fire detection probe (mounted over the test area) detects a flame. The hotplate in the Multiflash Automatic Flash Point - Universal Base Unit (34000-0) is equipped with a cut-out to protect against over-temperature.
The optional Cooling Boost Accessory (34008-0) allows the test area to be rapidly cooled in between tests by a blast of compressed air.
Ionization ring flash detector
Sweep arm setup tools
|TEMPERATURE RANGE||Ambient to 400 °C|
|POST TEST COOLING||Fan assisted or Forced Air (Requires 34008-0 Cooling Boost Module option)|
|POST TEST COOL-DOWN TIME (90°C TO 50°C)||9 minutes (Fan) 6 minutes (Forced Air)|
|IGNITOR||Quick-fit coaxial gas jet / electric hot wire, user-selectable (Electric hot wire pilot light for a gas jet)|
|TEMPERATURE PROBE||Class A Platinum Resistance Thermometer, °C or °F|
|FLASH / FIRE DETECTION||Ionization Ring|
|GAS SUPPLY (GAS JET IGNITOR)||Laboratory gas, 3 kPa (0.44 psi) maximum pressure|
|AIR SUPPLY (FORCED AIR COOLING||Dry compressed air, 670 kPa (100 psi) maximum pressure|
|VOLTAGE||110/120 V or 220/240 V, 50/60 Hz, (switchable)|
|SIZE (HXWXD)||330 x 470 x 420 mm|
|CCCN CODE||Tariff 90268020|
Draught Screen – Multiflash (Cleveland) - 34010-0
RS232 Cable (to connect instrument to PC) - 34003-0
Cooling Module - 34008-0
Hussami Kit (Cleveland) - 34301-0
Data Printer - 81002-3
Co-axial Ignitor (Multiflash PMCC, Cleveland) - 34100-004
Insulating Collets (pack of 2) - 34300-014
Ionization Ring (Multiflash Cleveland) - 34300-201
Cleveland Standard 80 ml (3 pack) - 99882-0
FLASHCHECK (pack of 3 x 80 ml) - 99880-0