Seta Multiflash Abel Flash Point Module – 34200-2
The automatic Abel module conforms precisely to national and international Abel flash point test methods. It comprises a water bath, sample cup and lid, temperature probes, flash detector, and a DIPS pod containing the dipping mechanism with co-axial gas/electric ignitor, and a stirrer.
The Abel module must be used with the Multiflash Universal Base Unit (34000-0). The base unit recognizes that the Abel module is connected and instantaneously sets up standard test parameters and calibration data. The Cooling Boost Module (34008-0) is required to test all types of samples.
Methods: IP 170, IP 492, BS 3900-A9 & BS 6664-2 & ISO 1523, BS EN ISO 13736 & BS 2000-170
- Full test method compliance
- Equilibrium and non-equilibrium tests
- Automatic set up of test parameters
- Gas or electric ignition
- Automatic heating control and flash detection
- Automatic test results with an end of test audible warning
- Centigrade or Fahrenheit temperature display
- Automatic barometric correction
The flash point test area consists of an outer liquid-filled bath surrounding an inner liquid-filled bath and a removable sample cup and lid. The outer bath is heated by the hotplate within the Universal Base Unit (34000-0), has a port for a temperature probe and connections for a Cooling Boost Module (34008-0), allowing circulation of cooling fluid via a cryostat for sub-ambient tests. The inner bath also has a provision for a temperature probe. The lid of the sample cup is equipped with a stirrer, sliding shutter, and ports for the flash detector and sample temperature probe.
The DIPS pod contains a stirrer drive, shutter actuator and ignitor dipping mechanism with a coaxial gas jet / electric hot wire ignitor.
The outer bath, inner bath, and sample temperature probes are class A PRTs, and flash detection is by a thermocouple.
Safety features include an initial dip to check that the sample is below its flash point at the start of the test and PRT checks to ensure that the correct probes are connected and are immersed in bath liquids or samples. When in search mode, the software limits the maximum temperature to 50 °C above “expected flash”. An over-temperature cut-out is mounted on the base flange of the outer bath.
Sample and bath PRTs
|TEMPERATURE RANGE||Ambient to 83 °C (-10 °C to 83 °C with cooling)|
|BATH||Liquid / Liquid|
|POST TEST COOLING||Fan assisted or circulating liquid / external cryostat (optional)|
|POST TEST COOL-DOWN TIME (90 °C TO 50 °C)||45 minutes (fan) 3 minutes (external cryostat)|
|IGNITOR||Quick-fit coaxial gas jet / electric hot wire, user selectable (electric hot wire pilot light for gas jet)|
|TEMPERATURE PROBES||Class A platinum resistance thermometers|
|STIRRER SPEED||30 rpm|
|GAS SUPPLY (GAS JET IGNITOR)||Laboratory gas, 3 kPa (0.44 psi) maximum pressure|
|COOLING SUPPLY||Water (5 °C and above) Water / ethylene glycol 50/50 (below 5 °C) 125 kPa (18 psi) maximum pressure|
|VOLTAGE||110/120 V or 220/240 V, 50/60 Hz, (switchable)|
|SIZE (HXWXD)||380 x 470 x 340 mm|
|CCCN CODE||Tariff 90268020|
Accessories, Spares, Calibration and Training
Data Printer – 81002-3
Cooling Module – 34008-0
RS232 Cable (to connect instrument to PC) – 34003-0
MTVM – Kerosine 500 ml – 99850-0
Cryostat (-20 °C) – 34006-3
Inner Bath Probe (Multiflash Abel) – 34200-004
Outer Bath Probe (Multiflash Abel) – 34200-003
PRT Probe (Multiflash Pensky-Martens) – 34100-002
Flash Detector (Multiflash PMCC, Tag, Abel) – 34100-001
Ignitor (Multiflash Abel) – 34200-005
Lazar Scientific is your reliable source for Flash PointTesting Equipment.