AvCount Sample Delivery Unit – SA1950-0
A compact system with a single push-button operation used to feed viscous samples into AvCount2 (SA1000-2) and AvCount Lite (SA1800-0). Suitable for use when viscosities exceed 68 mm2/s and up to 200 mm2/s.
Comprises a compressor and a sample pressurization chamber. A sample bottle is placed on the sample platform and the platform is raised into the pressurization chamber where it is pressurized to 3 bar by an integral compressor. The sample passes to the particle counter under pressure via the sample delivery tube. The delivery tube contains a non-return valve to prevent sample leakage when the sample delivery system is disconnected from the particle counter.
Safety devices include a pressure relief valve to protect against overpressure and a mechanical interlock prevents an operator from opening the pressurization chamber when it is pressurized.
Methods: ASTM D975, ASTM D7619, IP 565, DEF STAN 91-091, DEF STAN 91-86, ISO 4406
Supplied with power adaptors for UK, USA, and EU.
|MAXIMUM SAMPLE VISCOSITY||200 mm2/s|
|MAX BOTTLE SIZE||145 mm x 60 mm diameter|
|VOLTAGE||Pressure delivery unit: 24 Vdc, 2A (mains adaptor supplied 100 to 240 Vac, 50/60 Hz, 1A)|
|SIZE (HXWXD)||50 x 17 x 28 cm|
|CCCN CODE||Tariff 90318020|
Spares and accessories for AvCount Sample Delivery Unit - SA1950-0
SDU Delivery Hose - SA1950-003
SDU delivery hose for use with AvCount Sample Delivery Unit (SA1950-0).
SDU Power Supply 24 V - SA1950-002
SDU power supply 24 V for use with the AvCount Sample Delivery Unit (SA1950-0).